Ambient Light Readings
The KFM-1100 shutter speed can be selected in a range from as long as 30 minutes to as fast at 1/8000 of a second (This range is selectable in full stop, _ stop or 1/3 stop increments). The light reading is displayed on the meter’s LCD window as both digital and analog data. If you wish to select a different shutter speed after a reading is taken, the aperture reading will automatically change accordingly. The KFM-1100 can read a very wide range of light from the very low equivalent EV -2.0 to EV 19.9 (incident light based on ISO 100). The meter can also be set to a Cine mode to be used with Cine cameras and display light readings from 8 to 128 frames/sec.
Flash Light Reading
When using with flash, select Cord or Non-Cord depending on the shooting conditions. For Cord setting, connect the Flash sync-cord to the Sync Terminal in the front of the KFM-1100 then a light reading can be taken by simply pressing the Measurement button on the side of the meter. When using a non-cord set-up, pressing the measurement button sets the meter in stand-by mode. When the flash is fired manually, the meter takes a reading.
When taking readings in mixed flash and ambient lighting, the KFM-1100 can measure the flash and ambient light almost simultaneously. The percentage of flash light in the over-all exposure is then displayed in the analyze scale on the LCD. The analyze scale is divided into 4 sections, each section represents flash contribution of approximately 25% of the total lighting. For example, if 3 sections of the analyze scale are lit, the flash to ambient light ratio is approx. 75% flash to 25% ambient. After taking a reading, the shutter speed can be changed and the meter will recalculate the exposure. Setting a slower shutter speed would result in a greater ambient to flash ratio. Setting a faster shutter speed would result in a lesser ambient to flash ratio. The analyze scale helps you balance the 2 sources of lighting for more predictable results, so you can set the lighting and exposure to get the look you want.
The KFM-1100s calculation function can be used to average readings stored in memory or bias exposures toward shadow or highlights when taking reflected light readings. This allows for more accurate recording of shadow or highlight detail.
The KFM-1100 has memory slots to store 2 separate readings. Both stored readings and the current reading are displayed simultaneously in the analog scale. Memory markers also show the number of stored readings. Having readings stored in memory makes it easier to analyze the contrast range of your subject, the lighting ratio or viewing several separate readings in relation to one another.
The KFM-1100 has a brightness difference function that can be used to adjust lighting ratios or check for background light uniformity, very useful for product photography, portraits, etc. Checking for lighting uniformity across the entire seen is a critical use for light meters.Simply take a reading on your main reference point, then press the S/A/H button and then take reading on the other areas of the scene. The meter will display the brightness differences in term of +/- EV in 0.1 EV increments.
At any time you can set the display readings to your liking for: Selecting the shutter speed stops (1, _, 1/3 stops) Selecting the F no. Display mode (F number + 0.1 increment intermediate value display, or just F number direct reading display) Selecting the exposure correction value (EV – 10 to 10).
Type Hand-held Exposure meter for incident and flash light
Reception Method Incident-light readings
Receptors Incident- Spherical diffuser, 270° Rotating receptor head
Receptor Elements Silicon photocell
Measuring Modes AMBI- Ambient light
CORD- Flash w/sync cord
NON-CORD- Flash without sync cord
Measuring Range Ambient light EV +/- 2.0 to 19.9
Flash-Incident light reading +/- 0.1 stop
Flash for incident light F No. 1.0 to 128 + 0.9 Stop
For 40° Reflected light F No. 1.0 to 90 + 0.9 stop
For 5° Spot Viewfinder F No. 1.0 to 90 + 0.9 stop
Repeatability +/- 0.1 EV
F Number F No. 1.0 to 128 + 0.9 stops (0.1 stop incr.)
EV value -17 to 40.8 (0.1 stop incr.)
Ambient 30 minutes to 1/8000 sec.
Flash 1 to 1/1000 sec.
Frame Rate 8 to 128 frames/sec
ISO sensitivity 3 to 8000
Exposure Difference -10 to +10 (0.1 stop)
Analog Scale F No. 1.0 to 90 (1/2 Stop)
Analyze Scale Ratio between Flash and Ambient, 0 to 100% in 25% increments
Power one AA_Alkaline dry-cell battery
Battery Life Approx. 50 hours continuous reading in AMBI mode
Operation Temperature range 15 to 120° (-10 to + 50 C) with 85% relative humidity
Dimensions 2.25 in (57mm) W x 6.2 in (157mm) H x 1.0 in. (26mm) D
Weight 4.7 oz. (135g) w/o battery
Standard Accessories Spherical diffuser, neck strap, case
To accommodate the demands for an affordable, lightweight and digital ready light meter, Sekonic redesigned...
The versatility of today's cameras is expanding at an amazing rate. Digital SLR cameras...
The Ultimate Lighting tool, analyse any light source to reveal its colour spectrum, peaks...
Subscribe to our newsletter and always be the first to hear about what is happening.
|Melbourne||(03) 9672 2222|
|Sydney||(02) 8004 7945|
|Brisbane||(07) 3103 1429|
|Perth||(08) 6102 6382|
|Adelaide||(08) 7200 3773|
|Darwin||(08) 8911 1262|
|Canberra||(02) 6176 0125|
|Hobart||(03) 6107 9442|
|Auckland||(09) 887 0479|
© 2017 michaels camera video digital.